|Regulatory Requirement||Document Section|
|MDD Annex I (Essential Requirements) and Annex II, V, VI or VII for EU Declaration of Conformity
(depending on device classification)
|MDR Annex I (General Safety and Performance Requirements) and Annex IV (EU Declaration of Conformity)||All|
|MPDG §8, §96, §97||All|
|ISO 13485:2015, Sections 4.2.3||All|
This SOP describes how a medical device is brought to the market while observing applicable regulatory requirements, including the compilation of a Technical Documentation and issuing a EU Declaration of Conformity for the medical device.
1. Decision on New or Revised Medical Device
This process is initiated once the Management decides to place a new or updated medical device on the market. If applicable, a responsible Product Manager for a new device is appointed. Together with the QMO, s/he determines the classification of a new device as part of the MDD / MDR Classification document.
|Participants||Management, Product Manager, QMO|
|Input||Management decision on new device|
|Output||Completed medical device classification|
2. Application for Conformity Assessment by Notified Body
If necessary based on the medical device classification, a Notified Body is involved for the conformity assessment for the new device.
NOTE: a Notified Body must be involved in the conformity assessment of all medical devices except for class I.
|Input||Medical device classification|
|Output||Contract with Notified Body|
3. Compilation of Technical Documentation and Declaration of Conformity
The technical documentation for the medical device is compiled by following respective processes for development (e.g. software development, risk management, usability testing). The Product Manager is responsible to coordinate this process step with all relevant members of the organization. Documentation for different versions of the device is kept separate.
Prior to submission to the Notified Body for conformity assessment (if applicable), the QMO checks the documentation for completeness according to the stipulations listed below:
- MDD Essential Requirements (Annex I)
- MDD EU Declaration of Conformity (Annex II, V, VI or VII depending on device class)
- MDR General Safety and Performance Requirements (Annex I)
- MDR EU Declaration of Conformity (Annex IV)
After successful approval by the Notified Body, the Management signs the declaration of conformity. A product (version) is considered released and placed on the market with the release of its declaration of conformity.
The Product Manager keeps a list of current and previous products and product versions.
|Participants||Product Manager, QMO|
|Output||Approval by Notified Body
Signed EU Declaration of Conformity
Updated List of Medical Devices and Product Versions
4. Registration of Medical Device (UDI)
Country-specific requirements for registration are observed before offering a new medical device (version) in a market. For the European market, the Product Manager follows the requirements set out in Art. 29 and Annex VI MDR by registering the device’s Unique Device Identifier (UDI) in the European Database for Medical Devices (Eudamed).
For the implementation of product changes, the Product Manager applies for a new UDI-DI if the changes entail one of the following:
- A significant change based on the Change Evaluation List
- Changed intended use
- New product features
- Changes in product interoperability, new interface or connected systems
- Non-trivial change to the user interface
- New supported operating system
- New programming language
- New algorithm model
- Non-trivial change to the software or database architecture
The Product Manager assigns a new UDI-PI for modifications such as:
- Bug fixes
- Security patches
- User interfaces (limited to usability and excluding security improvements)
- The Basic UDI-DI (or: BUDI-DI) is assigned by UDI providers and identifies a group of medical devices with the same purpose, risk class, comparable design and manufacturing features.
- The UDI-DI is assigned by UDI providers for each medical device product type and is revised for any major modifications (substantial updates) to the device.
- The UDI-PI you can assign yourself as a manufacturer for every single product and is revised for minor modifications (non-substantial updates).
- Basic UDI-DI: Wrist-band health tracking device
- UDI-DI: different code based on color and packaging (white/blue/red; single-/double-pack)
- UDI-PI: different code based on series/batch/version, expiration date, etc.
NOTE: As of early 2021, the European Commission has designated four entities to issue UDI codes: GS1, HIBCC, ICCBBA and IFA. As long as Eudamed is not set up for registration, national registration requirements apply. For Germany, consider §96 and §97 of the MPDG. At OpenReg, we recommend obtaining a UDI already to be prepared for the regulatory transition.
NOTE: As part of the registration process, authorities in non-EU countries may require the documentation to be provided in other languages. Consider norms and regulations for the translation of medical device documentation.
|Input||Released Declaration of Conformity|
|Output||Updated or new UDI Completed device registration|
Template Copyright openregulatory.com. See template license.
Please don’t remove this notice even if you’ve modified contents of this template.